==========================================
 Tektronics 2445 scope PSU repair: part 3
==========================================

:Author: David Douard
:Category: Electronics
:Tags: Tektronics, 2445, analog scope, repair, test equipment
:series: Tek2445 PSU repair
:series_index: 3


After having mostly `destroyed the PSU of my Tek 2445
<{filename}/TeK2445_2.rst>`_, I've decided to be a bit less foolhardy
and I've subscribed to the `yahoo Tek group`_ and discovered that such
a failure has been reported there, also after a full recap of the
preregulator and regulator of the power supply.

I found that I was not so wrong when I built my completely guessed
LR1060.

It seems to be built on a 75Ω resistor. Mine was not so far, but I
decided to rebuild one with closer specifications, so I took a new 68Ω
resistor and I wrapped some 34 gauge wire. The result is a bit
cleaner, from:

.. image:: {filename}/images/tek2445/lr1060.jpg
   :alt: first attempt of a hand made LR1060 chocke.

to:
	 
.. image:: {filename}/images/tek2445/lr1060_v2.jpg
   :alt: a better hand made LR1060 chocke.
	 

But I was stuck because of my dead TR1050. Due to the lack of sucess
getting the main oscillator starts, I ended by suspecting TR1050
suffered when Q1050 exploded... And I was right: one of the two
primary coils was short.

I've found a reasonnably cheap replacement part on `QService
Electronics`_ shop. It took a bit long to arrive (12 days from Greece
to France, but to be honest, it seems the longest part of the transit
was in France), but it arrived. So yesterday evening I've been able to
replce it.


The TR1050 transformer
======================

This transformer is a small input transformer with 4 coils around a
common ferrite core:

.. image:: {filename}/images/tek2445/tr1050.jpg
   :alt: The TR1050 input transformer.

I was not sure of the specifications of each coil, since mine as (at
least on one coil) defective.  From the schematic, it looked natural
that both the primary windings should have the same charactritics. So
when my replacement transformer arrived, I took a few measurements.

The result is:

.. list-table:: TR1050 windings
   :widths: 15 10 30
   :header-rows: 1

   * - Winding
     - R
     - L
   * - 1-2
     - 4Ω
     - 6mH
   * - 4-5
     - 4.5Ω
     - 6.5mH
   * - 6-7
     - 0Ω85
     - ~ 10µH
   * - 6-7
     - 0Ω8
     - ~ 10µH

I've also quickly checked the voltage ratios using a 42kHz signal:

.. image:: {filename}/images/tek2445/tr1050_ratio_measurement.jpg
   :alt: Measuring the turn-ratios of the TR1050 transformer.

I use CH2 of my scope to probe the signal produced by the `HP890A4
<{filename}/hp8904a.rst>`_ linked to pins 1 and 2 (thus a primary
winding), and CH1 on a secondary winding (pins 6-7), which resulted
in:

.. image:: {filename}/images/tek2445/DS1Z_QuickPrint17.png
   :alt: Measuring the turn-ratios of the TR1050 transformer.
	 
.. image:: {filename}/images/tek2445/DS1Z_QuickPrint18.png
   :alt: Measuring the turn-ratios of the TR1050 transformer.

As one can see, the turn ratio seems to be 10. Both the 2 secondary
windings (5-6 and 7-8) which provide power for the preregulator
control (U1030) and the inverter drive (U1062, U1064 and U1066), are
identical to each other.

Testing the new T1050 transformer
=================================

After having soldered the "new" T1050 transformer, I followed one part
of the Power Supply Troubleshooting Procedure from the Service
Manual. In the diagram, there is a path in which there is:

- Unsolder and lift the end of W1060 near pin 7 of T1060;

- Connect the primary test load between the lifted end of W1060 and
  the sources of Q1060 and Q1070;

- Power up and check for 120V accross load (plus a 1Vpp triangle
  wave).

  The test load consist in:

.. image:: {filename}/images/tek2445/psu_test_load.png
   :alt: Primary Test Load.

So I built one and connected it:

.. image:: {filename}/images/tek2445/psu_with_test_load.jpg
   :alt: The PSU with the test load in place of T1060.

Was a little anxious when I switch the PSU on, but it did not
exploded. And the voltage across the dummy load was just fine!

So I resoldered W1060, reinserted all the jumpers (between A2 and A3),
plugged in a dummy load on J303 (unregulated 5V), a tried again. The
PSU was starting... it was not completely stable though, but I was
pretty sure this was due to the fact the load was not correct. So I
reinstalled the PSU in the scope, and gave it a try:


.. image:: {filename}/images/tek2445/back_from_the_death.jpg
   :alt: It's alive!

At last!


Summary
=======

I've spent a lot of time trying to understand and fix this PSU. I
think I've lost a lot of time and effort because I've been too hasty,
and I should have taken more time reading the service manuals. By the
way, the service manual for the Tek 2465A is a better source of
documentation, for example the question I had about the voltages with
respect to "REF" on the schematic is answered in this later manual:
there are indeed 2 references, as I suspected.

The main lesson from this is: never test a switching power supply
unloaded. I'm pretty sure I blew up the power transistors, the
transformer and a few other parts because I was switching the
*unloaded* PSU on and off. It's probably a lesson 101 on switching
power supplies... never too late to learn.

A the end, I've replaced many parts on my A3 preregulator board. Some
are probably not required:

- T1050,
- Q1060 and Q1070: I replaced with IRF820,
- Q1050: replaced with a IRFP450;the higher gate capacitance seems not
  to be problematic finally,
- Q1022, Q1030, Q1040 and Q1063: replaced with 2N4403 and BC337 (the
  transistors I found at my local components store)
- almost every capacitor,
- U1029 and U1040 optocoupler ,
- U1066: replaced by a DS0026CN bought on ebay
- CR1040: replaced by an unknown germanium "similar looking" diode,
- R1060 and R1070,
- R1069

So I'm done for now with this puppy. It should be completely checked,
but that will be another story.
  

Post Scriptum
=============

For the fun:

.. image:: {filename}/images/tek2445/side_by_side.jpg
   :alt: Side-by-side: DS1054Z vs. Tek2445
      
Despite the fact the Rigol DS1054Z has an incredible number of
features for which the Tek cannot compete, I still like this
old puppy a lot. Its front panel is much easier to use: every control is
directly accessible. And yet, I don't think the Rigol can measure both
the rise and fall time (with a decent precision) of a square wave:

.. image:: {filename}/images/tek2445/rise_and_fall.jpg
   :alt: Side-by-side: DS1054Z vs. Tek2445

Beautiful double time-base!


.. _`yahoo Tek group`: https://groups.yahoo.com/neo/groups/TekScopes/info
.. _`QService Electronics`: http://www.qservice.eu/
